EXERCISE MODES, APPLICATIONS, AND DIAGNOSTIC EVALUATION
In addition to providing bi-directional, bilateral, or unilateral exercise, the CES can create User Defined exercises. User Defined exercises can include an entire range of variations on standard exercises that will allow the clinician or coach to create an exercise for a specific need of the user.
Most conventional exercises start in the upstroke or upward movement of the bar except for the Squat which is assumed to start with downward motion first. The CES has the ability to reprogram the starting position of any exercise.
4.1 Exercise ModesThere are two separate modes of exercises, with 3 subfunctions, through which the CES controls the performance and data of an exercise. All five of these appear in the Exercise Modes box in the parameter screen.
4.1.1 Variable Velocity ModeVariable Velocity mode is a passive mode where the machine accommodates the forces exerted by the user throughout the range of motion, and at the same time restricts the speed of movement to a predetermined velocity pattern which may simulate the dynamics of real motion. This mode offers accommodating resistance during acceleration and deceleration as well as during isometric holds.
In Variable Velocity mode, the velocity or speed at which the bar is permitted to move is maintained at a pre-selected value, regardless of how hard the user pushes against it. In addition to maintaining a constant velocity, the ARIEL CES can be programmed to vary the velocity as a function of bar position in any pattern that the user may desire, thus the name Variable Velocity.
Variable Velocity mode exercise has a distinct advantage over resistance mode exercise: you don't need to know the individual's strength level in order to set up or begin exercising, nor is any adjustment necessary as the subject trains and increases in strength. The subject always pushes as hard as he or she can against the bar, and the computer automatically adjusts the resistance in order to maintain the desired bar velocity. In addition to maintaining a constant velocity, the Ariel CES can be programmed to vary the velocity as a function of bar position in any patter that you desire.
4.1.2 Variable Resistance ModeVariable Resistance is the mode most similar to traditional weight training. In the variable resistance mode, a level of force can be set which the user must exert, in order to move the bar. In addition to maintaining a constant resistance, the ARIEL CES can be programmed to vary the resistance as a function of bar position in any pattern that the user may desire, thus the name Variable Resistance. This feature allows the user to program greater or lesser amounts of weight, at any point in the motion, to accommodate for injuries and natural mechanical inefficiencies. It is well known feature of anatomy that the mechanical advantage of most muscles changes as the associated body joint is flexed or extended. The variable resistance mode allows an exercise to compensate for this change, becoming progressively easier or more difficult, and thus maintain a more constant muscular exertion through the entire range of movement.
The reporting changes in speed as a function of changing load is an excellent complementary information to the force/torque, curve the discovery of certain pathological cases, as well as talent recognition in youths and adults are only a few examples of the benefits of this mode.
4.2 SubfunctionsThe following three functions are all subfunctions of the Variable Velocity and Variable Resistance modes. Each subfunction can be used in either the Variable Velocity Mode or the Variable Resistance Mode.
4.2.1 Work TrainingWork Training is a velocity mode exercise with an additional feature. Rather than specifying the number of repetitions to be performed, work training specifies the total amount of work to be performed. Whether the user generates a great deal of work with only a few repetitions and greater force, or many repetitions with little force, a certain amount of work must be completed. Work is defined as the product of the force on the bar times the distance the bar is moved, and thus is proportional to the energy expended during exercise. Work training exercises, when performed at a continuous pace with a reasonably high amount of work, have a cardiovascular training effect as well as a muscular training effect.
4.2.2 Fatigue TrainingFatigue Training is a velocity mode exercise with an additional feature. Rather than specifying the number of repetitions to be performed, fatigue training specifies the degree of muscular fatigue to be achieved during exercise. fatigue is measured by comparing the average force for each repetition (upstroke portion) with the highest average force measured for all repetitions. This measurement is expressed as a percent of the highest average force, so that your fatigue level starts at 100%, and drops as your average force level drops. For example, a 75% fatigue exercise would continue until the average force produced in two consecutive repetitions was 75% or less of the average force for the strongest repetition. Fatigue mode can be viewed as a means to increase endurance, by gradually increasing the amount of repetitions that the user can do before falling below the fatigue floor. The fatigue mode is also an extremely efficient way of maintaining intensity in an exercise, as the user will need to maintain a certain level of intensity to continue the exercise.
4.2.3 Timed ExerciseTimed Exercise can be performed in either variable resistance or variable velocity mode. Rather than specifying the number of repetitions to be performed, timed exercise specifies the period of time. At the end of that time period that exercise will terminate regardless of the amount of work or repetition completed.
4.3 ApplicationThe application of all these exercise modes will become obvious to anyone who becomes familiar with the system. There is no replacement for experience in working with the system, however, there are some basic guidelines that serve as good starting points in setting levels on the CES for specific types of training. There are six basic categories that serve as outlines for the most common uses:
Set Duration for Muscular Endurance: 45 seconds - 3 minutes
The purpose of demonstrating this approach is to make the user understand that there is a very close relationship between the different modes. The information reported in each mode can also be used to tailor the program more accurately and more specifically to the needs of the exerciser.
4.4 Diagnostic EvaluationDiagnostic Evaluation is a feature that allows the CES to measure the user's maximum capabilities. These measurements are used in a variety of ways by the ARIEL CES. When diagnostics are being performed up to three values are measured and recorded:
When Diagnostics are performed in the manual mode, the values are only stored for as long as the same exercise is being performed. Diagnostics in the programmed mode will be stored in the memory of the program and can be updated from time to time. You may have only one set to Diagnostic values per each exercise.
4.5 Sampling Period for Rehab DataThe system allows 5 minutes or 250 repetitions, whichever comes first, for Rehab Sampling. During this period of time, the system collects all Rehab data on every repetition performed for statistical analysis and graphical presentation. The information in summarized in Rehab Table, Rehab Ratio, Stamina Report, Wave form, and the fatigue curves. The Sampling period does not effect any of the exercise ratios. An exercise can be performed for over 250 repetitions, but data will only be measured for the first 250 repetitions.